Flow diverters (FDs) are increasingly used to treat complex intracranial aneurysms, but preclinical studies that could guide clinical applications are lacking. The authors designed a modular aneurysm model in canines to address this problem.
Three variants of one modular aneurysm model were constructed in 21 animals. Sidewall (n = 5), curved sidewall (n = 5), and end-wall bifurcation (n = 7) aneurysms were treated with prototype 36-wire FDs. Four more end-wall bifurcation aneurysms were treated with prototype 48-wire lower-porosity FDs. Angiographic results postimplantation and at 3 months were scored with an ordinal scale. Animals were euthanized at 3 (n = 17) or 6 (n = 3) months, and the FD covering the aneurysm ostium was photographed to analyze metallic porosity and amount of neointima formation.
Straight sidewall aneurysms were better occluded than curved sidewall and end-wall bifurcation aneurysms at the 3-month angiography follow-up (p = 0.010). Flow diverters failed to occlude curved sidewall aneurysms (n = 0/5) and all but one (n = 1/7) end-wall bifurcation aneurysm. Angiographic results were no better (n = 0/4) using a 48-wire FD (p = 0.788). Branches jailed by the FD (n = 16) remained patent in all cases. Metallic porosity was decreased (p = 0.014) and neointimal closure of the aneurysm ostium was more complete (p = 0.040) in sidewall aneurysms than in curved or bifurcation variants of the model.
Flow diverters may succeed in treating straight sidewall aneurysms, but the same device repeatedly fails to occlude curved sidewall and end-wall bifurcation aneurysms. In vivo studies can be designed to test basic principles that, once validated, may serve to guide clinical use of new devices.
Abbreviations used in this paper:ASA = acetylsalicylic acid; CFD = computational fluid dynamic; FD = flow diverter; FSS = free segment of the stent.
Address correspondence to: Jean Raymond, M.D., Department of Radiology, Centre Hospitalier de l'Université de Montréal–Notre-Dame Hospital, 1560 Sherbrooke East, Pavilion Simard, Rm Z12909, Montreal, Quebec H2L 4M1, Canada. email: firstname.lastname@example.org.
Please include this information when citing this paper: published online May 4, 2012; DOI: 10.3171/2012.4.JNS111916.
CebralJRCastroMAAppanaboyinaSPutmanCMMillanDFrangiAF: Efficient pipeline for image-based patient-specific analysis of cerebral aneurysm hemodynamics: technique and sensitivity. IEEE Trans Med Imaging24:457–4672005
FiorellaDAlbuquerqueFCDeshmukhVRWooHHRasmussenPAMasarykTJ: Endovascular reconstruction with the Neuroform stent as monotherapy for the treatment of uncoilable intradural pseudoaneurysms. Neurosurgery59:291–3002006
FordMDLeeSWLownieSPHoldsworthDWSteinmanDA: On the effect of parent-aneurysm angle on flow patterns in basilar tip aneurysms: towards a surrogate geometric marker of intra-aneurismal hemodynamics. J Biomech41:241–2482008
HassanTAhmedYMHassanAA: The adverse effects of flow-diverter stent-like devices on the flow pattern of saccular intracranial aneurysm models: computational fluid dynamics study. Acta Neurochir (Wien)153:1633–16402011
KulcsárZErnemannUWetzelSGBockAGoerickeSPanagiotopoulosV: High-profile flow diverter (silk) implantation in the basilar artery: efficacy in the treatment of aneurysms and the role of the perforators. Stroke41:1690–16962010
LylykPMirandaCCerattoRFerrarioAScrivanoELunaHR: Curative endovascular reconstruction of cerebral aneurysms with the pipeline embolization device: the Buenos Aires experience. Neurosurgery64:632–6432009
MutFAppanaboyinaSCebralJR: Simulation of stent deployment in patient-specific cerebral aneurysm models for their hemodynamics analysis. Proceedings of the Summer Bioengineering Conference of the American Society of Mechanical EngineersMarco Island, Florida2008
SzikoraIBerenteiZKulcsarZMarosfoiMVajdaZSLeeW: Treatment of intracranial aneurysms by functional reconstruction of the parent artery: the Budapest experience with the pipeline embolization device. AJNR Am J Neuroradiol31:1139–11472010