Ahmed K. Toma, Andrew Tarnaris, Neil D. Kitchen and Laurence D. Watkins
Managing symptomatic ventriculoperitoneal shunts with no clear evidence of shunt malfunction either clinically or radiologically can be a difficult task. The aim of this study was to assess intracranial pressure (ICP) monitoring as a method of investigating shunt function.
The authors performed a retrospective analysis of 38 continuous ICP monitoring procedures done in patients with ventriculoperitoneal shunts and suspected shunt malfunction.
Thirty-eight procedures were performed in 31 patients between January 2005 and October 2008. Sixteen recordings were normal, 6 revealed overdrainage or low pressure, 11 indicated underdrainage or high pressure, and 5 showed variable shunt function. Based on the findings after 20 procedures (53%), patients were treated conservatively: 4 by readjusting the valve setting and 16 by referral to the headache neurologist for medical treatment. Forty-five percent of the conservatively treated patients improved. Surgical exploration was undertaken following 18 procedures (47%); 72% of the surgically treated patients improved.
Continuous ICP monitoring using an intraparenchymal probe is a safe and effective method of investigating adult hydrocephalus.
Ahmed K. Toma, Andrew Tarnaris, Joan P. Grieve, Laurence D. Watkins and Neil D. Kitchen
In this paper, the authors' goal was to compare the artifact induced by implanted (in vivo) adjustable shunt valves in spin echo, diffusion weighted (DW), and gradient echo MR imaging pulse sequences.
The MR images obtained in 8 patients with proGAV and 6 patients with Strata II adjustable shunt valves were assessed for artifact areas in different planes as well as the total volume for different pulse sequences.
Artifacts induced by the Strata II valve were significantly larger than those induced by proGAV valve in spin echo MR imaging pulse sequence (29,761 vs 2450 mm3 on T2-weighted fast spin echo, p = 0.003) and DW images (100,138 vs 38,955 mm3, p = 0.025). Artifacts were more marked on DW MR images than on spin echo pulse sequencse for both valve types.
Adjustable valve–induced artifacts can conceal brain pathology on MR images. This should influence the choice of valve implantation site and the type of valve used. The effect of artifacts on DW images should be highlighted pending the development of less MR imaging artifact–inducing adjustable shunt valves.